Visit us on the web at for more information: • Price Quotations • Drivers· Technical Specifications. Manuals and Documentation . Model System DMM Instruction Manual Rev. D. Manual Type: Primary User; Part Number: D. Release Date: Sep 17 View and Download Keithley DMM instruction manual online. 5 function autoranging digital multimeter. DMM Measuring Instruments pdf manual.
|Published (Last):||19 May 2007|
|PDF File Size:||15.93 Mb|
|ePub File Size:||8.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
As far as the local switch goes, the Keithley part number isbut unless you have a donor unit it would be pretty hard to find.
Kerry D. Wong » Blog Archive » Keithley Firmware/Calibration Data Backup
These programs are listed in Table Enter the number 35 by pressing the “3” and “5” but- tons. The in- strument will return to the ekithley operating state. Use a standard BNC con- nector to make the connection to the Model HP Progamming Example— Enter the following statements into the computer to obtain and display the machine status word UO. I am having some trouble getting the two to communicate with each other.
To modify the address, key in a new value with the numeric data buttons. The IFC command can be sent by typing in the following statement: If an additional instruction manual is required, order the manual package Keithley Part Number With this program, the ,eithley will indicate whether or not a specific reading falls within a prescribed range.
Enter the number 31 by pressing the ” 3 ” and ” 1 ” but- tons. Do not use zero to cancel the offset in this procedure. The Q command is in the following form: Table shows the current output for each resistance range. HP statements have a one or three digit argument that must be specified as part of the statement.
It is important to note that the increased display range does not increase the maximum allowable input level to the instrument.
The current HI limit will be displayed. The Model cable is identical to the Modelbut is 1. Each of these indicators is briefly described below. Thus, manua is recommended that the selected function button be pressed periodically.
The secondary function of this but- ton is to enter the number 0. When the Model completes the reading, it will tng- ger the Model to go to the next channel.
The current primary address wUl be displayed. Model instruction manual with schematics, Rev.
196 System DMM Instruction Manual | Keithley
Additional connections, which are not kdithley on the diagram, will also be necessary to apply signal inputs to the scanner cards, as well as for the signal lines between the scanner and the Model The controller sets ATN true.
In the continuous trigger mode, the instrument takes a continuous series of readings.
If the self-test is successful, the J byte in the UO status word will be set to nanual. The current primary address will be displayed.
Keithley 196 DMM Manuals
Select a range consistent with the expected resistance or use autorange. Perform the follow- ing steps mahual use this program: Calibration standard for next units: LSN will be off when the unit is in the listener idle state. Also, tests on the dis- play and memory are performed. This increased display range ensures that the dynamic measurement range of die instrument is not reduced when using a zero baseline value.